A monitor for an intravenous pump audio alarm. The client needs to be alerted by a vibrating pager as she is deaf. The pump has no alarm output other than an audio signal. A system has been designed and built that can distinguish the pump alarm in relatively high ambient noise and reject other events such as knocking or tapping on the microphone sensor.
Some people with e.g. cerebral palsy are unable to use physical controllers such as buttons and joysticks but are still able to make intentional hand movements. HandShake uses a pair of BBC micro:bits to enable triggers to be sent to switchable communication software to enable speech to be composed. A trigger is sent to the communications device when the participant moves a hand above an adjustable threshold of acceleration.
The client with motor neuron disease has inadequate strength in his hands to operate his mattress elevator to allow him to get out of bed unaided, or to summon assistance. Foot operated switches were attached to the side of his bed.
Description : Ear switch as an assistive communication and control device / gaming switch and HCI
The Client wears a positive air pressure mask at all times. This makes it difficult for people to hear what she says.
This is a handle which is designed to clip onto a BT 4600 DECT Phone to aid grip. It is for those with weaker hands and/or limited mobility, and means that if grip is lost, the phone remains in the hand rather than dropping.
This is a handle which is designed to clip onto an Alcatel 1066 Mobile Phone to aid grip. There is a rigid and collapsible version of the handle available.
The lady had suffered from sepsis and, as a consequence, had lost both hands and both feet. The NHS were able to provide a wheelchair and later prosthetic legs but had little to offer in the short term to provide hand functions. We were able to provide basic implements to enable her to feed herself, brush her hair, and to write birthday cards and sign cheques.
Child-sized portable voice amplifier for a child with a very soft voice
Our client was having trouble communicating with those around her. Her low vision and other disabilities made talking and using equipment difficult. The solution, specially developed by Remap, was a dedicated speech communication device.
Aluminium mount with detent pin Tom’s speech therapists were keen that he should have his Communicator with him at all times. The normal arrangement of a shoulder strap meant that he could swing the device, which while he enjoyed doing it, it hardly facilitated its use, and was distracting both to him, his teacher and […]
Identified need The client has no fine motor control of his hands but can use arms and/or head. He is confined to bed and feels insecure at night as he is not able to alert a carer in another room or part of the house. Description This unit is based on the Wilko Standard Portable […]
Client has a communication device consisting of a power pack and touch screen. This needs to be mounted on her wheelchair Brackets were constructed to fit a box section tube on the wheelchair
The client, a young person, is unable to speak and was being taught to use a communication aid. He needed to use this aid both in his wheelchair and his armchair but the manufacturer did not have any suitable means of fixing it within his limited reach. A suitable platform to hold the device was […]
This child has severe cerebral palsy. His only method of communication was through a speech synthesising computer and he needed the next level of computer to continue progress but this was unable to be mounted on his powered wheelchair. An aluminium tube was mounted on the lower left side of the wheelchair with provision for […]
Some 12 years ago, the client had a laryngetomy operation. She has also developed severe arthritis. The action required to cover the stoma in her throat with one finger is causing painful RSI. A ‘finger’ was made from Perspex rod. The end was hinged and covered with a soft rubber compound to give a positive […]